TY - JOUR

T1 - Optimization of the integration time of pulse shape analysis for dual-layer GSO detector with different amount of Ce

AU - Yamamoto, Seiichi

PY - 2008/3/21

Y1 - 2008/3/21

N2 - For a multi-layer depth-of-interaction (DOI) detector using different decay times, pulse shape analysis based on two different integration times is often used to distinguish scintillators in DOI direction. This method measures a partial integration and a full integration, and calculates the ratio of these two to obtain the pulse shape distribution. The full integration time is usually set to integrate full width of the scintillation pulse. However, the optimum partial integration time is not obvious for obtaining the best separation of the pulse shape distribution. To make it clear, a theoretical analysis and experiments were conducted for pulse shape analysis by changing the partial integration time using a scintillation detector of GSOs with different amount of Ce. A scintillation detector with 1-in. round photomultiplier tube (PMT) optically coupled GSO of 1.5 mol% (decay time: 35 ns) and that of 0.5 mol% (decay time: 60 ns) was used for the experiments. The signal from PMT was digitally integrated with partial (50-150 ns) and full (160 ns) integration times and ratio of these two was calculated to obtain the pulse shape distribution. In the theoretical analysis, partial integration time of 50 ns showed largest distance between two peaks of the pulse shape distribution. In the experiments, it showed maximum at 70-80 ns of partial integration time. The peak to valley ratio showed the maximum at 120-130 ns. Because the separation of two peaks is determined by the peak to valley ratio, we conclude the optimum partial integration time for these combinations of GSOs is around 120-130 ns, relatively longer than the expected value.

AB - For a multi-layer depth-of-interaction (DOI) detector using different decay times, pulse shape analysis based on two different integration times is often used to distinguish scintillators in DOI direction. This method measures a partial integration and a full integration, and calculates the ratio of these two to obtain the pulse shape distribution. The full integration time is usually set to integrate full width of the scintillation pulse. However, the optimum partial integration time is not obvious for obtaining the best separation of the pulse shape distribution. To make it clear, a theoretical analysis and experiments were conducted for pulse shape analysis by changing the partial integration time using a scintillation detector of GSOs with different amount of Ce. A scintillation detector with 1-in. round photomultiplier tube (PMT) optically coupled GSO of 1.5 mol% (decay time: 35 ns) and that of 0.5 mol% (decay time: 60 ns) was used for the experiments. The signal from PMT was digitally integrated with partial (50-150 ns) and full (160 ns) integration times and ratio of these two was calculated to obtain the pulse shape distribution. In the theoretical analysis, partial integration time of 50 ns showed largest distance between two peaks of the pulse shape distribution. In the experiments, it showed maximum at 70-80 ns of partial integration time. The peak to valley ratio showed the maximum at 120-130 ns. Because the separation of two peaks is determined by the peak to valley ratio, we conclude the optimum partial integration time for these combinations of GSOs is around 120-130 ns, relatively longer than the expected value.

KW - Decay time

KW - Depth-of-interaction

KW - Gamma detector

KW - Integration time

KW - PET

KW - Pulse-shape analysis

UR - http://www.scopus.com/inward/record.url?scp=40849116042&partnerID=8YFLogxK

UR - http://www.scopus.com/inward/citedby.url?scp=40849116042&partnerID=8YFLogxK

U2 - 10.1016/j.nima.2008.01.084

DO - 10.1016/j.nima.2008.01.084

M3 - Article

AN - SCOPUS:40849116042

SN - 0168-9002

VL - 587

SP - 319

EP - 323

JO - Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment

JF - Nuclear Instruments and Methods in Physics Research, Section A: Accelerators, Spectrometers, Detectors and Associated Equipment

IS - 2-3

ER -